The following is an abstract for the selected article. A PDF download of the full text of this article is available here. Members may download full texts at no charge. Non-members may be charged a small fee for certain articles.
Optimum Photometric Range of Ultraviolet Spectrophotometers
Volume 19, Number 1 (Jan. 1965) Page 32-33
Gridgeman (1) assumed that a spectrophotometer would be limited in photometric performance by the readout mechanism. The absorbance, A, is related to the transmittance, T, by:A = - log T, then dA/dT = - (log e)/T.