The following is an abstract for the selected article. A PDF download of the full text of this article is available here. Members may download full texts at no charge. Non-members may be charged a small fee for certain articles.
A Digital Recorder for Mass Spectra
Volume 12, Number 2 (May 1958) Page 40-46
As a mass spectrum is scanned, the maximum value of each peak is stored in a capacitor. Then the difference between the stored voltage and the lesser voltage seen after the scan has proceeded down the side of the peak operates a peak detector, which in turn initiates measurement of the stored peak height by a digital voltmeter. At the same time the mass is determined by measuring the ion accelerating voltage with a modified self balancing potentiometer equipped with a shaft position digitizer.