The following is an abstract for the selected article. A PDF download of the full text of this article is available here. Members may download full texts at no charge. Non-members may be charged a small fee for certain articles.
Improved Fluorescence Rejection in Measurements of Raman Spectra of Fluorescent Compounds
Volume 42, Number 2 (Feb. 1988) Page 248-254
Kamogawa, Keiji; Fujii, Takashi; Kitagawa, Teizo
Practical ways to achieve fluorescence rejection were developed by using the technique of Raman difference spectroscopy and a gated diode-array detector. In order to improve the signal-to-noise ratio, we had to circumvent the problem presented by the fact that the sensitivities of each element of the diode-array detector are not always uniform and linear—a problem which results in an apparent noise in the observed spectra. Therefore, in this study we tried to separately observe a Raman-plus-fluorescence spectrum (ST) and a pure fluorescence spectrum (SF), with a high S/N ratio, and to obtain a Raman spectrum (SR) as their difference. With a cw Ar+ laser, ST and ST were excited at different wavelengths but recorded in the same wavelength region. This procedure was easy and quite successful for compounds, such as methyl orange and methylene blue, that have a Raman-to-fluorescence intensity ratio (IR/IF) of greater than 0.01. With a mode-locked and Q-switched Nd:YAG laser, ST and SF were obtained from different timings of the detector gate (6 ns), i.e., −6 to 0 ns for ST and 3 to 9 ns for SF. The temporal discrimination between Raman and fluorescence photons combined with the Raman difference technique significantly improved the 532-nm excited Raman spectrum of the tetraphenylporphyrin solution. This method is applicable to compounds with an IR/IF of smaller than 0.01. The time-correlated single photon counting method with a time-to-amplitude converter was improved, to enable it to work with a high repetition rate, and was applied to a Rhodamine 6G solution, which was also examined by the gated Raman difference technique. Advantages and problems associated with each technique and appropriate methods for a given sample will be discussed.