The following is an abstract for the selected article. A PDF download of the full text of this article is available here. Members may download full texts at no charge. Non-members may be charged a small fee for certain articles.
Volume 13, Number 4 (Aug. 1959) Page 110-110
Baker, Drexel W.
It is common practice to use some sort of lens system between the arc-spark stand and the spectrograph in spectrochemical analyses. In many instances the lens closest to the arc gap receives considerable spatter from the arc and gradually deteriorates. A metal shield, shown in Figure 1, limits the unobstructed path between the arc gap and the lens to the cone required by the optical arrangement. This reduces the number of spatter particles hitting the lens, especially the destructive heavy particles which require trajectories that are cut off by the shield. The device also aids in keeping dust off the lens.